Pointe Scientific, Inc, Canton, MI is initiating a nationwide recall of all size kits of Liquid Glucose Hexokinase Reagent catalog number G7517. The reagents have been found to fail linearity at >200mg/dL that results in inaccurate glucose values above this range.
Distributors and testing laboratories who have received the Pointe Scientific, Inc Liquid Glucose Hexokinase Reagent (G7517) which is being recalled should destroy remaining inventory.
The recall includes the following lot numbers:
Lot # 823901 (all batches) Expiration: 2010-02
Lot # 826801 (all batches) Expiration: 2010-03
Lot # 829401 (all batches) Expiration: 2010-04
Lot # 831502 (all batches) Expiration: 2010-05
Catalog #s: G7517-120 (97 kits), G7517-500(129 kits), G7517-1L (31), 8-G7517-120 (15 vials), 8-G7517-500 (21vials), 3-G7517-L (28L), 7-G7517-500 (5 vials). 7-G7517-1000 (3bottles)
The firm voluntarily recalled the products after confirming a reported assay failure in Lot 826801. FDA has been apprised of this action.
Testing laboratories should consider all test results obtained with the lot numbers listed above to be questionable. The laboratory should inform the patient’s attending physician, and determine, with their input, whether confirmation of the previous test results will be required.
To date, the firm is not aware of any injuries or inaccurate results being reported out.
The Pointe Scientific, Inc Liquid Glucose Hexokinase Reagent (G7517) was distributed to medical device distributors and/or drop-shipped to testing laboratories nationwide by courier delivery service. Product involved in the recall can be identified by any of the lot numbers listed above appearing on the Pointe Scientific, Inc kit content label, vial label or bulk container label.
Pointe Scientific, Inc has notified its distributors and drop-ship customers by e-mail, fax, telephone and/or certified mail and is arranging for replacement of all recalled products.
Distributors and testing laboratories with questions may contact the company at 1-800-445-9853.